IEC 61010-2-091:2018 PRV
Pre release version
Safety requirements for electrical equipment for measurement, control and laboratory use - Part 2-091: Particular requirements for cabinet X-ray systems
IEC 61010-2-091:2018 specifies particular safety requirements for cabinet X-ray systems, which fall under any of categories a), b) or c) below.
a) Electrical test and measurement equipment
This is equipment which by electromagnetic means tests, measures, indicates or records one or more electrical or physical quantities, also non-measuring equipment such as signal generators, measurement standards, power supplies for laboratory use, transducers, transmitters, etc.
NOTE 1 This includes bench-top power supplies intended to aid a testing or measuring operation on another piece of equipment. Power supplies intended to power equipment are within the scope of IEC 61558 (see 1.1.2 h)). This standard also applies to test equipment integrated into manufacturing processes and intended for testing manufactured devices.
NOTE 2 Manufacturing test equipment is likely to be installed adjacent to and interconnected with industrial machinery in this application.
b) Electrical industrial process-control equipment
This is equipment which controls one or more output quantities to specific values, with each value determined by manual setting, by local or remote programming, or by one or more input variables.
c) Electrical laboratory equipment
This is equipment which measures, indicates, monitors, inspects or analyses materials, or is used to prepare materials, and includes in vitro diagnostic (IVD) equipment.
This equipment may also be used in areas other than laboratories; examples include selftest IVD equipment to be used in the home and inspection equipment to be used to check people or material during transportation.
This second edition cancels and replaces the first edition published in 2012. It constitutes a technical revision.
This edition includes the following significant changes from the first edition, as well as numerous other changes:
- The scope of the document has been clarified and limited to equipment below 500 kV.
- Additional marking requirements for X-ray generating assemblies have been added. (5.1)
- Requirements for high-voltage cables used in the X-ray assembly have been added. (6.5)
- Insulation requirements have been added. (6.7)
- Temperature requirements for beam-limiting devices have been added. (10.3)
- Clarification has been provided on and , and test methods. (12)
- Requirements for have been modified, taking into account functional safety standards. (15)
- Requirements for reasonably foreseeable misuse have been clarified. (16)
- Risk assessment has been made mandatory for specific aspects. (17)
|Publication type||International Standard|
|TC/SC||TC 66 - Safety of measuring, control and laboratory equipmentrss|
|ICS||19.080 - Electrical and electronic testing
71.040.10 - Chemical laboratories. Laboratory equipment
|File size||1283 KB|
The following test report forms are related:
Share your publications
Learn how to share your publications with your colleagues, using networking options.
Our prices are in Swiss francs (CHF). We accept all major credit cards (American Express, Mastercard and Visa), PayPal and bank transfers as form of payment.
Keep in touch
Keep up to date with new publication releases and announcements with our free IEC Just Published email newsletter.
Contact customer services
Please send your enquiry by email or call us on +41 22 919 02 11 between 09:00 – 17:00 CET Monday to Friday.