Evaluation and routine testing in medical imaging departments - Part 3-5: Acceptance tests and and constancy tests - Imaging performance of computed tomography X-ray equipment
IEC 60601-2-44 and this document
• defines the essential parameters which describe the performance of CT SCANNERS with regard to image quality, RADIATION OUTPUT and PATIENT positioning; the list of parameters to be tested can be found in section 4.3,
• defines the methods of testing the essential parameters, and
• evaluates compliance with the tolerances of the parameters SPECIFIED by the ACCOMPANYING DOCUMENTS.
The methods defined in IEC 60601-2-44 and this document rely on non-invasive measurements, using appropriate test equipment, performed during or after installation. Signed statements covering steps in the installation procedure can be used as part of the ACCEPTANCE TEST report.
This document applies to ACCEPTANCE TESTS and CONSTANCY TESTS on a CT SCANNER. The aim of the ACCEPTANCE TESTS is to verify compliance of the installation or MAJOR SERVICE ACTION with specifications affecting the image quality, RADIATION OUTPUT and PATIENT positioning. The CONSTANCY TESTS are performed to ensure that the functional performance of EQUIPMENT meets ESTABLISHED CRITERIA and to enable the early recognition of changes in the properties of components of the EQUIPMENT, and to verify compliance with specifications affecting the image quality, RADIATION OUTPUT and PATIENT positioning.
This document also contains requirements associated with ACCEPTANCE and CONSTANCY TESTING for the ACCOMPANYING DOCUMENTS of the CT SCANNER.
This document does not apply to
• aspects of mechanical and electrical safety, and
• aspects of mechanical, electrical and software performance, unless they are essential for performing the ACCEPTANCE TESTS and CONSTANCY TESTS, and are directly affecting image quality, RADIATION OUTPUT and PATIENT positioning.
IEC 61223-3-5:2019 cancels and replaces the first edition published in 2004, and the second edition of IEC 61223-2-6 published in 2006. This edition constitutes a technical revision.
This edition includes the following significant technical change with respect to the previous edition and to IEC 61223-2-6:
1) modification of the RADIATION protection and control;
2) modification of the acceptance testing;
3) introduction of constancy testing.
Share your publications
Learn how to share your publications with your colleagues, using networking options.
Our prices are in Swiss francs (CHF). We accept all major credit cards (American Express, Mastercard and Visa), PayPal and bank transfers as form of payment.
Keep in touch
Keep up to date with new publication releases and announcements with our free IEC Just Published email newsletter.
Contact customer services
Please send your enquiry by email or call us on +41 22 919 02 11 between 09:00 – 17:00 CET Monday to Friday.